Electromagnetic flow converter for combination with OPTIFLUX and WATERFLUX flow sensors. KROHNE 03/ – – TD IFC R05 en. The documentation is only complete when used in combination with the relevant documentation for the . 2. The high-performance solution. IFC is the only electromagnetic flow converter with 3 x diagnostics of instrument, application and .
|Published (Last):||25 May 2009|
|PDF File Size:||17.25 Mb|
|ePub File Size:||20.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Simulation of the flow fraction for partly filled pipes. Display of the actual flow fraction for partly filled pipes.
C checks in progress Test mode of the device. Dimensions and weights For detailed information refer to chapter “Dimensions and weights”. Product Liability And Warranty F configuration also when changing modules Invalid configuration: Or the demanded krohme rate is too high.
Information Concerning The Documentation If installed and operated in accordance with these operating instructions, it will rarely present any problems. Pulse And Frequency Output A dialogue appears in the display.
Inputs And Outputs Save the data to the backplane again Service. Signal converter for electromagnetic flowmeters. All operating data and output values can be adjusted.
Krohne IFC 300 Signal Converter for Electromagnetic Flowmeters
The collection of personal data such as names, street addresses or e-mail addresses in the kdohne documents is always on a voluntary basis whenever possible. Connection diagram for measuring sensor, field housing 1 2 3 4 5 6 48 Electrical terminal compartment in housing of the signal converter for signal and field current cable. The operating point of the 4 optical keys is located directly behind the glass pane.
Crimp wire end ferrules onto both conductors.
This warning refers to the immediate danger when using this device in a hazardous atmosphere. Full pipe electrode has no contact with medium. In hazardous areas, grounding is used at the same time for equipotential bonding. Make setting, see Fct. DS type A Max. Simulation of the level for partly filled pipes. The manufacturer accepts no liability for the consequences of misuse by the operator.
The correct position is indicated by the steady bright red LED of the interface, as soon as the red LED and the IR-Sensor below the keys on the display are roughly one above the other, see Fig.
PROFINET I/O option for electromagnetic flowmeters | KROHNE Group
Construction of signal cable B 1 Stranded drain wire for the inner shield 101. All-round signal converter matches almost any application requirement Reliable measurements with extensive 300 of device and application Wide range of communication options including PROFINET The IFC is a very complete signal converter, featuring a wide range of variants and options to match almost any application requirements in process industries.
Aluminium and aluminium sheet, partially polyester-coated Option Versions C and F: Display When Setting Parameters, 4 Lines If this limit occurs sporadically in processes with air pockets, solid contents or low conductivity, then either the limit has to be increased or a pulse filter used so as to quell the error messages and also reduce the measurement errors.
Noisy flow indic and flow pipeline is flowing full, C 1. IFC Signal Converter. The cables are factory connected and labeled as follows.
This offers great time and cost benefits with regard to procurement, engineering, operation and servicing. Note the key function within and between the columns.
Description Of The Cg Number Measuring sensor DN [mm] [inch] Min.